How do I refill my Ammonia Cap?
- Remove the end cap of the electrode. Half fill the cap with a pipette or other tool.
- Tap the side of the cap gently to remove any air bubbles that could interrupt conduction.
- Screw the cap into the electrode until the pH meter touches the Teflon membrane. The membrane should not tear or stretch more than the minimum. This means that there will be a thin layer of solution between the membrane and probe.
- You can then connect the electrode directly to the TRUEscience SMART cap.